Apr. 27 , 2019
One is the failure of the Oximax SpO2 Sensor itself, and the other is caused by external factors of the probe.
The failure of the probe itself mainly focuses on the disconnection of the cable, the burning of the luminous tube and the photoelectric tube, or the decrease of the efficiency of transmitting and receiving. Under normal circumstances, it can be judged by static measurement, and then specifically excluded for the fault point. Faults caused by external factors of the Adult Finger Clip SpO2 Sensor mainly focus on noise interference, patient jitter or severe keratinization of the probe, probe emission, unclean portion of the receiving part, and light leakage. As long as the source of the interference is found, the probe is cleaned, the connection port is reinforced, and the correct use can be eliminated on site. The current monitoring instruments generally have an alarm function, and the fault code or prompt when the alarm is one of the shortcuts to find the cause of the fault.
Sep. 20, 2019Find here details of Spo2 Sensor on our website. Today we would like to talk about the range of body temperature fluctuations of normal ... View More
Sep. 16, 2019What should you pay attention to when selecting the ECG lead? Today the professional Philips ECG Cable Manufacturer would like to... View More